Understand how to set the controller add-on.
Note: To receive your controller add-on code please contact your Kuntze representative.
The Controller add-on offers a concentration-based controller with two independently configurable control relays. When you activate the add-on, the main menu shows an additional symbol: CONTROL
Navigate to the Controller add-on: Main menu > CONTROL
S1/S2 CONTROL define the set point, the controller type, and its corresponding parameters
S1/S2 CONFIG define the controller output via the relays (i.e. pulse-pause or pulse-frequency)
In the submenu RELAY you can define the acting direction, such as whether dosing should happen when the measured value is above or below the set point. Here you can also turn off relays if you want to use the current outputs as controller output.
In the submenu TIMING you can define a start delay and S1/S2 dosage-check times
Helpful article: Activate Add-Ons - Krypton® DIS